In case of direct measurement in the field the auger hole method according to Hooghoudt can be applied.Following this method the determination of the permeability to water of a soil takes little time (by comparison to other methods), requires a limited number of instruments.
The principle of this method is quite simple. A hole is bored in the ground to a certain depth below the groundwater level and after a time the water in the hole will rise to the said water level. The water is consequently bailed out and the time it takes for a new water level to establish itself is recorded. With the help of formulae and/or nomogrammes this rate of water rise can be translated to the average water permeability factor of the different strata of soil.
The depth of the bore hole is dependent upon the groundwater level and the thickness and the permeability of the successive layers in the ground profile.
The standard set (09.01.SA) is suitable for measurements to a depth of 2 meter. Basically it contains: an Edelman- and a Riverside auger with an upper part and an extension rod, a bailer, measuring tape with holder and float, a filter and a stopwatch. The complete set can be transported in the field in a strong carrying bag.
The extendable set (09.01.SB) is suitable for measurements to a depth of 5 meter. This set includes the same items as the standard set, however the 1 meter long filter has been replaced by an extendable filter with a total length up to 5 meter (also including extension rods for the auger).
Product code: 0907
Guelph constant head permeameter for measurements to a depth of 15-75 cm, complete with field...More information